Home › Forum › Support › CPM not available on website
My Kit 1 unit has stopped collecting the radiation readout on website sensor ID: 13000214
I can see the reading on the LCD and hitting the local IP displays a CPM value so I am confused as the issue with the portal recording the radiation readings.
Is there anything that can be done to troubleshoot?
Can you post what you see on all the screens of your KIT1 unit?
Here is the screen as well as the web output.
The browser screen capture was taken a couple of mins after the physical screen.
Seems it is up and running: https://uradmonitor.com/tools/dashboard-04/?open=13000214
The unit is online.
What seems to be the problem?
The OP’s unit shows radiation level 0 all the time:
That’s not normal, since there is proper high voltage.
Thanks Wolferl for clarifying this for me
Problem fixed. The unit was flagged on the server for its radiation readings, possible something bad happened in the past.
Also another fix for the unit in Moscow – a DIY KIT1 unit, class 0x13: Link:https://www.uradmonitor.com/?sensor=cpm&open=13000200
It was sending the “Tube type parameter” – index “10” (hex) defined in Exp protocol Link:https://github.com/radhoo/uradmonitor_kit1/blob/master/code/misc/expProtocol.h as “0x1” .
This did not work because of the 0x , therefore your unit’s factor was set to 0, making sievert/h outputs 0. Code on server was modified to check and convert such inputs. It is now all set.